Pro3dure GR-18.1 IB is a biocompatible 3D printing resin for the generative production of orthodontic ”indirect bonding trays“. The post cured material is flexible, transparent and has a high tear strength. This allows the orthodontist to precisely place all brackets in one operation.
COMPATIBLE WITH DLP/LCD/SLA 3D PRINTERS
|Density:||ca. 1.1 g/ml|
|Viscosity||ca. 2.4 Pa s|
on postcuring unit)
|Hardness:||ca. 80 Shore A|
|Storage||between 64°F/18°C and 83°F/28°C|
|Tensile Strenght||n.a./n.a.*||6.2 MPa||DIN 53504|
|Elongation at break||n.a./n.a.*||50%||DIN 53504|
|Shore Hardness||n.a./n.a.*||40D||ISO 7619-1|
|Viscosity (23°C)||n.a./n.a.*||2.5 Pa s||DIN 53019-1|
|Biocompatibility||Irritation and delayed allergies||comply||ISO 10993-10|
How to use
GR-18.1 bottles should be well shaked before use (fig 1).
Make sure that GR-18.1 material is temperature adjusted up to 23 °C to 30 °C.
Carefully pour GR-18.1 into the vat of the image projection unit (fig 2).
Bubbles can be removed with a cleaned spatula or by a recoater routine.
If it is possible, always store a bottle GR-18.1 in your production unit in order to avoid temperature differences during refilling.
For the build parameter adjustment please refer to the machine data sheet.
After the build process is finished a direct post treatment is recommended.
If this cannot be guaranteed leave the produced objects in the liquid GR-18.1 resin.
After cleaning of the parts with isopropanole ≥ 97 % (approx. 3-5 min. in an ultrasonic bath) the objects are postcured in an adequate light curing unit in a protective gas atmosphere.
The dental objects generated out of the generative resin GR-18.1
can be repaired with GR-18.1 resin.
Impurity due to operation mistakes cannot be excluded. With respect to the low viscosity of the resin it is possible to filtrate the GR-18.1 . It is recommended to filtrate and stir up the resin on a regular base (fig 3).
To avoid bubbles let GR-18.1 rest for 1 hour before usage.
To avoid detrimental effects on material quality do not expose the liquid material to irradiation under any circumstances.
Deviations from the described manufacturing process may lead to different mechanical and optical properties of the GR-18.1 material.
Ensure personal protective gear during processing.
Caution: Polymerised resins are chemically resistant -avoid stains on clothing.
Avoid any contact with skin and eyes. In case of accidental contact, rinse with adequate running water, consulting a doctor if necessary.
- The lot number and the expiration date are indicated on each bottle of GR-18.1 . In case of claims please always indicate the lot number of the product. Do not use the product after expiry of the best before date.
The company I.D.M di Marin Davide is not a manufacturer of medical materials or devices. The material sold on this site is neither produced nor controlled by the company I.D.M. di Marin Davide, who in no case can be held responsible for any damage caused by use or application of the material, which must only be used on suitable machines and by specialized personnel, exclusively for the purpose indicated, according to the manufacturer’s instructions. For resins with European certification, the properties of the finished product vary according to the entire printing and post-curing process; refer to the manufacturer for any information about it.