Dental Bleach 3D printing resin is a ceramic-filled resin for printing temporary crowns and bridges with Bleach color according Vita scale.
It has high hardness 90 Shore D and high tensile strength, stable to mechanical stress and chemicals. Recommended for printing of temporary crowns and bridges.
Low odor and absence of harmful monomers allows to print in the laboratory.
Russian certification. Product without european certification, to be used only for models.
COMPATIBLE WITH MAIN DLP/LCD 3D PRINTERS
3D printing settings on most widespread 3D printers
Harz Labs Dental Bleach 3D printing resin in the version for SLA 3D printers, such as Form1 / 2 and Peopoly, is available upon request. Write to us at: firstname.lastname@example.org.
3D printing resin Harz Labs Dental Bleach characteristics
|Viscosity (Brookfield)||at 25°C||1,2-1,5 Pa.s|
|Hardness (Shore D)||90-92|
|Tensile strenght||60 N/mm2|
|Elongation at break||3-4%%|
UNIQUE MECHANICAL PROPERTIES
Very high hardness, tensile strength, abrasion and scratch resistance due to ceramic fillers.
Resins are odorless and cause no irritation, so you can work with natural ventilation without specialequipment.
Depending on postcuring A1 or A2 color is achievable. It is stable for long time even in contact with aggressive agents.
Produced from bio-compatible raw material used in photopolymer teeth fillings. Meets standards ISO 10993.
Resin is stable to solvents, aggressive inorganic agents like acid and alkalis.
Pigments and fillers are well stabilized, no sedimentation and flocculation occur, no foam appears.
Correct post-processing process for temporaries
To remove excess resin, wash the 3D printed part in an ultrasonic bath with the TEK1960 * solvent cleaner for 2 cycles of 2-3 minutes.
Do not soak in acetone. Do not leave the 3D part soaking in the liquid. Use a soft brush if necessary.
Dry with compressed air.
Heat for 10-20 min in oven at 60 * C
Perform post-curing in a oven with turntable and 36-48 W UV LED lamps for 30-60 minutes
Polish and apply protective varnish (eg Optiglaze)
*We do not recommend the use of denatured or isopropyl alcohol (IPA). It is very aggressive on the polymerized resin and it could weaken the structure of the 3D part. It is also flammable and potentially harmful to people and the environment.
Store the resin in a dry, cool, shaded place out of direct sunlight.
The company I.D.M di Marin Davide is not a manufacturer of medical materials or devices. The material sold on this site is neither produced nor controlled by the company I.D.M. di Marin Davide, who in no case can be held responsible for any damage caused by use or application of the material, which must only be used on suitable machines and by specialized personnel, exclusively for the purpose indicated, according to the manufacturer’s instructions. For resins with European certification, the properties of the finished product vary according to the entire printing and post-curing process; refer to the manufacturer for any information about it.